In Situ TEM Biasing & Heating
under controlled electrical and thermal environment.
Lightning Application Fields
Dedicated or simultaneous biasing & heating studies
Nano-scale investigations of nano-electronic devices & materials
E-field induced dynamics of piezoelectric nanoparticle
Domain evolution in ferroelectric materials
Ferroelectric materials are characterized by the existence of spontaneous electric polarizations at a temperature well below Curie temperature. In a small area, the polarizations may share a same direction and form the so-called ferroelectric domain. The spontaneous polarization can be reversed by applying an external electric field exceeding the coercive field. Investigating the ferroelectric properties at both temperature and external electric field is important for applications such as data storage and optical frequency converters. Here we present a recent experiment performed using the Lightning D9+ (now HB+) demonstrating the capability of delivering both high electric field and stable temperature. Simultaneous electric field and heating is also possible, however, not shown below. The material under investigation is BZT-0.5BCT, which has a Curie temperature around 90 °C and a coercive electric field of 2~4kV/cm.
In Situ TEM Analysis of Organic–Inorganic Metal-Halide Perovskite Solar Cells under Electrical Bias
Abstract Changes in the nanostructure of methylammonium lead iodide (MAPbI3) perovskite solar cells are assessed as a function of current–voltage stimulus by biasing thin samples in situ in a transmission electron microscope. Various degradation pathways are identified both in situ and ex situ, predominantly at the positively biased MAPbI3 interface. Iodide migrates into the positively biased charge transport layer and also volatilizes along with organic species, which triggers the nucleation of PbI2 nanoparticles and voids and hence decreases the cell performance.
Up to 8 contacts for simultaneous biasing & heating
The Lightning Nano-Chips features 4-point-probe method to accurately control biasing and heating and retrieve meaningful data. The design of the Nano-Chip sustains the highest fields and temperatures to be reached (dedicated or simultaneous), enabling the characterization of today’s and tomorrow materials..
Introducing the Nano-Chip to the microscope
Made from titanium for its optimal mechanical stability, the Lightning double tilt Sample Holder is the critical element connecting the Nano-Chip with the microscope: designed for the highest flexibility it is compatible with all TEM techniques and features a wide tilt range to be able to look at samples from most orientations.
Total control over the biasing & heating environment
For biasing experiments a source measuring unit (SMU) is required to precisely source voltage or current and simultaneously measure voltage and/or current. The majority of SMUs are compatible with the Lightning system and our preferred supplier/model is Keithley 2450. The Keithley 2450 can conveniently be controlled via the new Impulse SW. Different Keithley models or other can be controlled via the manufacturer’s own software.
Heating experiments are managed via Impulse software.
A seamless integrated system for total control
The new Impulse software provides the user with full control over temperature and electric field. It enables faster experimental setup, easy customization of the workspace and easy monitoring of the experiment
400 kV/cm @ RTBiasing
300 kV/cm @ 900 °CSimultaneous Heating&Biasing
Sample preparation with conventional techniques
The sample preparation techniques used for preparing traditional TEM samples including lamellas, nanowires and particles are suitable for the Nano-Chip. FIB lamellas are the most commonly used sample for biasing experiments and DENSsolutions in conjunction with some close academic partners have developed a unique FIB workflow using a customised FIB stub specifically designed for the Nano-Chip. This process significantly reduces the total workflow time and makes the success in transfer much higher. Additional methods such as micro-manipulators are suitable for sample preparation onto the Nano-Chip.
Some great work from our customers
Frequently Asked Questions
What is the size of sample recommended for biasing experiments?
What preparation methods other than FIB could be usable for biasing experiments?
What is the homogeneity of the electrical field?
What TEM pole-pieces are compatible?
What really sets the 50V / 100V limit?
- To avoid the electric failure between connection pins in the vacuum. (Electric sparks)
- To avoid breakdown of SiNx at high electric field (the limit is lower at elevated temperature).
Be aware, the specified voltage is not the breakdown voltage of our system, but in fact to ensure a low leakage current. Therefore, in reality one could go much higher than the specified numbers if the experimental details allow.
Can my sample survive the electrical loading induced?
Download the Lightning Brochure
For more information on features and specifications.
Lightning – In situ TEM Characterization of Electrical Properties of Semiconductor Nanowires
Lightning – In situ atomic-scale observation of electrochemical delithiation in a working solid state battery
Feel free to contact us with any further questions.
Request a Demo or Quote
Request a quotation or demonstration at your lab.